Rated Burst Pressure 22 atm
Ultra-low tip entry profile (0.41 mm/ 0.016″)
- Low diameter growth between nominal and rated burst pressure (lowest among current generation of NC balloons)
- Low entry & crossing profile
- Kink resistant shaft
- 5F compatible
- Highly flexible distal shaft
- PTFE coated proximal shaft
Diameter growth between Nominal Pressure (NP) and Rated Burst Pressure (RBP)
Bench test results may not necessarily be indicative of clinical performance. Test performed by and data on file at Sahajanand Medical Technologies Pvt. Ltd. Testing performed on 2.50x15mm of each leading brand- Pantera LEO (Biotronik), Sapphire II (Orbusneich) , NC Trek (Abbott), Accuforce (Terumo) and Pipit NC of Sahajanand Medical Technologies
|Balloon material||Nylon 12|
|Balloon fold||3-fold memory|
|Tip entry profile||0.016″(0.41mm)|
|Shaft diameter||Proximal 1.95F, Distal 2.67F|
|Rapid exchange length||25cm|
|Markers on the proximal shaft||90cm and 100cm|
|Maximum recommended guidewire||0.014″ (0.36mm)|
|Guide catheter compatibility||5F Compatible|
Pipit NC is a registered trademark of Sahajanand Medical Technologies. Specifications are subject to modification, revision and improvement.
Caution: This product is intended for use by or under the direction of a physician. Prior to use refer the “Instructions for use” provided inside the product carton (when available) for more detailed information on indications, contraindications, warnings, precautions, and adverse events.
Information contained herein for distribution outside the U.S., Japan & France.
Illustrations are artist’s representations only and should not be considered as engineering drawings or photographs. Photos on file at SMT.
NC Trek is a registered trademark of Abbott Cardiovascular Systems. Pantera LEO is a registered trademark of Biotronik Inc. Sapphire II is a registered trademark of Orbusnieich & its affiliates. Accuforce is a registered trademark of Terumo.